Reliance on the use of cellular telephones and similar wireless communication devices has become pervasive in our society. From initially being able to send and receive only voice communications, today’s mobile wireless communication devices are equipped with text messaging capabilities, Internet access, digital cameras, voice recorders, global positioning system (or GPS) functions, customized software applications, portable music players, and more.
When used in health care facilities such as hospitals, these functional and convenient radiofrequency wireless communication technologies contribute to the electromagnetic environment of the facility. There is therefore a need to mitigate potential risks of electromagnetic interference between cell phones and similar wireless communication devices and sensitive medical equipment in clinical health care environments. Health care facilities now need to develop policies that balance a need for effective communication with wireless communication devices, while minimizing risks of electromagnetic interference with critical medical equipment.1 As cell phones and similar mobile wireless communication devices and medical equipment technologies evolve, these polices will require regular review.
The objective of this report is to provide an overview of Canadian tertiary care hospital policies on wireless communication device use. For the purposes of this report, “wireless communication devices” include two-way radios, cell phones, “smart phones” (e.g., iPhone, Android), as well as wireless personal digital assistants (or PDAs), which among other capabilities can store and/or transmit voice calls, emails, and data messages. Only policies developed and/or revised within the last three years have been included.
It is not intended that the findings of this Environmental Scan provide a comprehensive review of the topic. Results are based on communication with key informants from the provinces. This report is based on information gathered as of September 2011.
Of the 10 Canadian provinces surveyed, responses were received from eight. Responses from more than one respondent were sought from the larger provinces of Quebec, Ontario, Alberta, and British Columbia. Respondents represented individual tertiary care hospitals, as well as provincial or regional-level health authorities, depending on at what level of the health care system the policies were developed.
Nine respondents indicated that they are currently revising their wireless communication devices policy, and were not able to release the draft document details. Four respondents shared their current policies, which had been revised within the last three years. These policies do not ban the use of wireless communication devices in their health care facilities. Rather, they allow for:
- unrestricted but respectful and privacy-conscious use in designated areas of the hospital, such as waiting areas, lounges, cafeterias, and private offices, where there is minimal medical equipment
- limited use (use within certain designated distances of medical equipment)
- prohibited use in highly instrumented patient care areas.
Due to the increased functionality of wireless communication devices, including digital cameras and voice recording functions, issues relating to patient and staff privacy and confidentiality have also been addressed in recently developed health care facility policies.
The following table summarizes the currently active (i.e., non-draft) wireless communication devices policies provided by survey respondents.
Table 1: Summaries of Wireless Communication Device Policies
The policy applies to all RF transmitting devices physically located in any of the hospital’s campuses, whether they belong to staff, patients, or visitors. The policy defines RF transmitting devices as any device emitting RF energy of sufficient power to transmit signals over controlled frequencies (e.g., cellular telephones, many PDAs [e.g., BlackBerry], wireless local area networks, wireless video cameras, pagers, and walkie-talkies). It is the policy to control the use of RF transmitting devices in all patient care and diagnostic areas to ensure that applicable technology can be used safely, while preserving patient privacy and minimizing patient care area disruptions. The policy is intended to balance patient privacy and safety needs with the benefits of incorporating emerging mobile technologies into patient care.
Two-way radios: Only Emergency Services (fire/police), hospital security, and patient transport personnel may use, as required, keeping as far away from medical equipment as is practical
(6 to 8 metres). Under emergency conditions, two-way radios may be distributed to selected staff users.
Receive-only and two-way pagers: May be used by physicians, staff, patients, and visitors without any restrictions.
RF transmitting devices may be used in non-patient care areas of the hospital, providing that doing so does not violate others’ need for quiet and privacy, and being courteous and respectful of others; or does not violate any patient’s rights with respect to their privacy and security of their health information; or does not violate any staff member’s privacy by taking a photo without their consent.
Appropriate signs limiting or prohibiting the use of RF transmitting devices are posted outside all restricted areas.
Any medical device malfunction that may have been caused by interference from an RF transmitting device must be reported immediately to Biomedical Engineering, who will also notify the Risk Manager/Shift Manager, who will follow established procedures.
RF devices taken into a patient’s bed space must be treated similarly to other patient equipment, in that the user must perform hand hygiene after using the device and before touching the patient.
A patient and visitor brochure on the use of cell phones was developed and made available.
|The policy applies to the use of wireless communication devices such as, but not limited to, cellular phones, pagers, PDAs, or two-way radios by patients, visitors, and all hospital personnel (collectively referred to as “all”). Recognizing that effective and timely communication is essential to the delivery of proper health care, and normal operation of any health care facility, the use of wireless communication devices for mission-critical purposes is allowed within the delineated guidelines.
Through appropriate entry signage, all individuals carrying cellular phones and other RF-emitting devices are requested to turn these devices off (note: Not in “standby” mode, in which transmission still occurs periodically).
Signage throughout the hospital indicates where cellular phone use is allowed or restricted.
Green cellular phone signs identify low-risk areas in which cellular phone use is permitted: These are defined as zones with few medical devices or minimal possibility of RF interference and include lobbies, the cafeteria, waiting areas, private offices, or other areas as designated by the green cellular phone signs.
Use of cellular phones is not permitted in high-risk areas designated by red cellular signs: These are defined as areas where a patient may be connected to electromedical equipment (e.g., cardiac monitors, hemodynamic monitors, medical telemetry), any other highly instrumented areas (e.g., clinical laboratories), patient rooms, or other areas as designated by the red cellular phone sign.
|Separate policies are in place for staff use and public use of wireless communication devices. “Restricted areas” for both the staff and public policies are identified as the following heavily instrumented areas:
Wireless communication devices are defined as those devices which intentionally transmit information throughout the air, and are divided into two categories:
Special Application wireless communication device use may be authorized by Clinical Engineering and other appropriate management, when there is a requirement, following appropriate risk analysis. Two-way radios are only for use by appropriately trained facility staff and emergency service personnel such as fire, police, and paramedic personnel. General contractors or third parties need authorization from Facility Management for each of the facilities’ sites.
Public policy: The purpose of the policy is to enhance patient safety by managing wireless communication devices that can interfere with proper operation of certain medical equipment, as well as balance obligations to ensure appropriate safety standards are met and to recognize the public’s continually increasing use of wireless communication devices.
Wireless communication devices are defined as those devices which intentionally transmit information throughout the air. Examples of wireless communication devices include cellular telephones, “smart phones” (i.e., Blackberry, Palm, iPhone, etc.), two-way pagers, two-way radios, wireless web-enabled laptops, or tablets and FRS.
The public may use their wireless communication devices throughout the facilities, except in restricted areas where they must be turned off (i.e., powered off, and not left in “stand-by” mode). Appropriate signs shall be posted at the entrances to restricted areas. In patient care or waiting areas, the vibration or “quiet” feature of the phone shall be used. All usage, including photographs or video recording for wireless communication devices having these capabilities, shall be in accordance with existing facilities’ policies of privacy, respectful environment, and confidentiality. An education booklet about the wireless communication policy has been developed for staff and public, and shall be available in all patient care areas.
For both staff and public policies: If Clinical Engineering staff become aware of medical equipment whose function has a reasonable probability of being adversely affected by wireless communication devices, the equipment shall be tagged accordingly, clearly indicating a minimum separation distance that shall be maintained between the device and any wireless communication device. If interference between a wireless communication device and a piece of medical equipment occurs or is suspected, the wireless communication device should be turned off immediately or removed from the area, and Clinical Engineering staff are to be appropriately notified.
|Wireless communication devices can be used in any area of the hospital. Increasing the distance between a wireless communication device and potentially susceptible medical equipment greatly reduces the risk of EMI.
Low-power wireless devices (< 1watt and unlicensed), defined as cellular phones, “smart phones” (e.g., Blackberry, iPhone, Palm), Androids (mobile devices), paging transmitters, medical telemetry transmitters, RFID (e.g., Pharmacy), FRS, or two-way radios (< 0.5 watts) must be kept at least one metre away from susceptible medical equipment (labelled as such by the Biomedical Engineering Department).
High-power wireless devices (1-5 watts and licensed), defined as GMRS two-way radios (used by maintenance, ambulance, fire, police, taxis, construction crews), must stay at least 3 metres away from susceptible medical equipment (labelled as such by the Biomedical Engineering Department).
Any observed malfunction of a medical device shall be reported to Biomedical Engineering.
Wireless communication device etiquette guidelines must be followed, including being mindful of surroundings and speaking quietly without use of abusive and foul language in the presence of others, and ensuring the protection of patient privacy.
A patient information sheet was developed to educate the public, and a staff information sheet was developed to educate staff, on the recently revised policy.
EEG = electroencephalogram; EM = electromagnetic; EMI = electromagnetic interference; FRS = family radio service; GMRS = general mobile radio service; ICU = intensive care unit; MRI = magnetic resonance imaging; PDA = personal digital assistant; RF = radio frequency; RFID = radio frequency identification device.
The majority of the Canadian hospitals surveyed in each of the ten provinces indicated that they are currently revising their wireless communication device policies. Those facilities that have revised their policies within the last three years indicated that use of cell phones and similar devices is not completely banned; however, usage is limited to within certain designated distances from medical equipment, or prohibited in areas that have medical equipment. Generally, respectful cell phone use is permitted in areas of the hospital such as waiting areas, lounges, private offices, and cafeterias having minimal medical equipment. Issues relating to privacy concerns and confidentiality have also been addressed in recently developed health care facility policies.
Complementing this Environmental Scan are two CADTH Rapid Response reports from 2010 pertaining to cellular telephone interference in hospitals and wireless device use in hospitals. These reports are available free of charge on the CADTH website.2,3
- Guidance on electromagnetic compatibility of medical devices in healthcare facilities [Internet]. Arlington (VA): Association for the Advancement of Medical Instrumentation; 2010. Report No.: AAMI TIR18:2010. [cited 2011 Nov 10]. Available from: http://marketplace.aami.org/eseries/scriptcontent/docs/Preview%20Files/T...
- Cellular telephone interference in hospitals: safety and guidelines [Internet]. Ottawa: Canadian Agency for Drugs and Technologies in Health; 2010 Nov 24. [cited 2011 Sep 22]. (Rapid response report). Available from: http://www.cadth.ca/media/pdf/K0282_Cell_Phones_Hospitals.pdf
- Wireless devices in hospitals: safety and guidelines [Internet]. Ottawa: Canadian Agency for Drugs and Technologies in Health; 2010 Dec 2. [cited 2011 Sep 22]. (Rapid response report). Available from: http://www.cadth.ca/media/pdf/htis/dec_2010/K0286_Wireless_Devices_Hospitals.pdf
Cite as: Loorand-Stiver, L.Canadian Hospital Policies on Wireless Communication Device Use [Environmental Scan issue 31]. Ottawa: Canadian Agency for Drugs and Technologies in Health; 2011.
CADTH takes sole responsibility for the final form and content of this environmental scan. The statements and conclusions in this environmental scan are those of CADTH.
Production of this report is made possible by financial contributions from Health Canada and the governments of Alberta, British Columbia, Manitoba, New Brunswick, Newfoundland and Labrador, Northwest Territories, Nova Scotia, Nunavut, Prince Edward Island, Saskatchewan, and Yukon. The Canadian Agency for Drugs and Technologies in Health takes sole responsibility for the final form and content of this report. The views expressed herein do not necessarily represent the views of Health Canada or any provincial or territorial government.
Disclaimer: The Environmental Scanning Service is an information service for those involved in planning and providing health care in Canada. Environmental Scanning Service responses are based on a limited literature search and are not comprehensive, systematic reviews. The intent is to provide information on a topic that CADTH could identify using all reasonable efforts within the time allowed. Environmental Scanning Service responses should be considered along with other types of information and health care considerations. The information included in this response is not intended to replace professional medical advice nor should it be construed as a recommendation for or against the use of a particular health technology. Readers are also cautioned that a lack of good quality evidence does not necessarily mean a lack of effectiveness, particularly in the case of new and emerging health technologies for which little information can be found but that may in future prove to be effective. While CADTH has taken care in the preparation of the report to ensure that its contents are accurate, complete, and up to date, CADTH does not make any guarantee to that effect. CADTH is not liable for any loss or damages resulting from use of the information in the report.
Copyright: This report contains CADTH copyright material. It may be copied and used for non-commercial purposes, provided that attribution is given to CADTH.
Links: This report may contain links to other information available on the websites of third parties on the Internet.
Canadian Agency for Drugs and Technologies in Health
(CADTH)600-865 Carling Avenue,
Ottawa, Ontario K1S 5S8